Waveform: ZOLL Rectilinear Biphasic.TM
Defibrillator Charge Hold Time: 30 seconds
Energy Selection: Automatic pre-programmed
selection (Adult Mode: 120J, 150J, 200J;
Pediatric Mode: 50J, 70J, 85J).
Patient Safety: All patient connections are
Charge Time: Less than 10 seconds with new
Electrodes: ZOLL CPR-D•padz, pedi•padz® II
or stat•padz® II.
Built-in Defibrillator Self Test: Included
CPR: Metronome Rate: Variable 60 to 100 CPM
Depth: 3/4” to 3”; 1.9 to 7.6 cm.
Defibrillation Advisory: Evaluates electrode
connection and patient ECG to determine if
defibrillation is required. Shockable Rhythms:
Ventricular fibrillation with average amplitude
>100 microvolts and wide complex ventricular
tachycardia with rates greater than 150 BPM for
adults, 200 BPM for pediatrics. For ECG Analysis
Algorithm sensitivity and specificity, refer to AED
Plus Administrator’s Guide.
Patient Impedance Measurement Range: 0 to
Defibrillator: Protected ECG Circuitry
Display Format: Optional LCD with Moving Bar
Size: 2.6” x 1.3”; 6.6 cm x 3.3 cm
Display Sweep Speed: 25 mm/sec; 1”/sec
Battery Capacity: Typical new (20°C) = stand-by
life with batteries installed: 5 years (300 ±5
continuous shocks: 250 ±5 shocks before “change
battery” indicator and 50 ±5 shocks after “change
battery” indicator) or 13 hours of continuous
monitoring time. For AED Plus units running
software version 5.32 or higher, replace batteries
every 5 years, or if unit prompts. For earlier
versions, please replace batteries every 3 years.
PC Minimum Requirements for Configuration and
Patient Data Recovery: Windows® 98, Windows®
2000, Windows® NT, Windows® XP, IBM
compatible PII with 16550 UART (or higher)
computer. 64MB RAM. VGA monitor or better.
CD-ROM drive. IrDA port. 20MB disk space.
Size: (H x W x D) 5.25”x 9.50” x 11.50”;
13.3 cm x 24.l cm x 29.2 cm
Weight: 6.7 lbs.; 3.1 kg
Power: User Replaceable Batteries. 10 -Type 123A
Photo Flash lithium manganese dioxide batteries.
Device Classification: Class II and internally
powered per EN60601-1
Design Standards: Meets applicable requirements
of UL 2601, AAMI DF-39, IEC 601-2-4, EN60601-1,
Operating Temperature: 32° to 122°F; 0° to 50°C
Storage Temperature: -22° to 158°F; -30° to 60°C
Humidity: 10 to 95% relative humidity,
Vibration: MIL Std. 810F, Min. Helicopter Test
Shock: IEC 68-2-27; 100G
Altitude: -300 to 15,000 ft.; -91m to 4573m
Particle and Water Ingress: IP-55.
Shelf Life: 5 years
Conductive Gel: Polymer Hydrogel
Conductive Element: Tin
Packaging: Multilayer foil laminate pouch
Impedance Class: Low
Cable Length: 43 in (1.1 m)
Sternum: Length: 6.1 in (15.5 cm); Width: 5.0
in (12.7 cm); Length, conductive gel: 3.5 in (8.9
cm); Width, conductive gel: 3.5 in (8.9 cm); Area,
conductive gel: 12.3 sq in (79.0 sq cm)
Apex: Length: 6.1 in (15.5 cm); Width: 5.6 in (14.1
cm); Length, conductive gel: 3.5 in (8.9 cm); Width,
conductive gel: 3.5 in (8.9 cm); Area, conductive gel:
12.3 sq in (79.0 sq cm)
Complete assembly: Folded Length: 7.6 in (19.4 cm);
Folded Width: 7.0 in (17.8 cm); Folded height: 1.5
in (3.8 cm)
Design standards: Meets applicable requirements
of ANSI/AAMI/ISO DF-39-1993.